Payment & Shipping Terms:
thermocouple temperature sensor,
ntc probe temperature sensor
High Precision NTC Temperature Sensor For Refrigerator Temperature Test
1. High precision high stability
2. Quick temperature response
3. Resistant to heat shock
4. Moisture resistant
5.Excellent quality and high stability by high technical mass production
Temperature test in kinds of air-condition,refrigerator,hot boiler,microwave oven.
2.0 Structure And Dimensions
2.1 Standard Configurations And Dimensions
3．0 Part Number Code
×× × ××× × × ××××
① ② ③ ④ ⑤ ⑥
Tolerance of R90
4.0 Performance Parameter
test method:90℃±0.1℃.(90℃±0.1℃ in the oil.)
4.2Thermal time Constant(τ)
test method:In the water, When the products be put into the water at 25℃ from the water at 50℃, Thermal time constant is the time that cost by the resistance becoming to be the value at34.2℃.
requirement:3mw/℃,3mW/℃(min, in air)
test method(The product will be join with the following circuit at 25±0.5℃ in still air.
4.4 Max.rated power
80℃ is the maximum temperature that the product is heat by itself.
test condition:At DC 500v.
requirement:There shall be no evidence of damage during the test.
test condition:AC 1500V for a One second,Permissible Current 5mA.
4.7Max. Permissible Current
test condition:The current which can be applied to the sensor for rised upper limit of operating temp. range at 25℃ in still air.
4.8 Temperature -40--105℃
|Testing item||Requirement||Testing method and condition.|
Dielectric withstanding voltage and insulation sistance not change.
There shall be no evidence of damage during the test.
-30℃,10min Room temperature 10min 100℃, 30min [in coolant ] [in water ]
Placed for 500hours Applting DC 1 mA at RT.
|Heat resistance test||
Charged 500hours at 100±2℃ in air.
|Cold resistance test||
Charged 500hours at －30±2℃ in air.
|Bias humidity test||500hrs.Placed 500hours at 55±2℃ 90%～95%RH.|
Lead wire crimping strength
1 min load with 3kg.
|Drop test||From height 1m drop on concrete floor 10 times.|
one time: 1 min.
Direction and time:X,Y and Z direction for 2hrs.
180°for 10 times bending the part between wire and epoxy.
6.0 Method of the stockpile
6.1 The height of stacks products cannot overbalance four boxes,in the stackpile and the course of transport.
6.2 The packing products of inside must placed desiccant.Allowance used the angone way to transport,but must prevent therain and the snow direct or indirect shower bath and damnification of these products.
6.3 The products are stockpiles candition of the tencperature is about -10℃/+40℃,the opposite humidity not bigger too 80%,acidic and alkalescent of substance,erode gas or radiant point are not should enclose the products.
Contact Person: Anna